Recently, I have seen many users on the Internet asking about the humidity of the rapid temperature change test box. In fact, it is a headache for this device, because the device is an environmental device. When the test box is used for a long time, it will naturally Some malfunctions occurred or the personnel were not working properly. The humidity problem mentioned above is actually related to the use of a long time or improper personnel. Today, the Fast Temperature Change Chamber Factory introduces its problems and solutions.
When the damp heat test is performed, the actual humidity will reach 100% or the actual humidity will be very different from the target humidity. The value is much lower. The former phenomenon may be caused by the drying of the gauze on the wet bulb sensor. This is the sink that needs to check the wet bulb sensor. Whether there is water shortage in the water, because the water level in the water tank is automatically controlled by the water level controller, it is necessary to check whether the water supply system and work are properly supplied.
It is also possible that wet gauze is used for a long period of time or the purity of the water supply, which will make the gauze harden so that the gauze can not absorb moisture and dry. At this time, it is only necessary to replace or clean the gauze to release. The latter phenomenon is that the humidification system does not work. It is necessary to check whether there is a certain amount of water in the water supply system of the humidification system, and whether the water level control for controlling the water level of the humidification boiler is normal, and whether the water level in the humidification boiler is normal. If all of the above is OK, then check the electrical control system and the manufacturer to perform the repair.
For the above-mentioned humidity problems and solutions, I don't know if you have a clear question about the humidity of the Rapid Temperature Change Chamber?
Copyright © AI SI LI (China) Test Equipment Co., Ltd. All Rights Reserved | Sitemap